- Set Impact Twin "stand alone Clock" to ADAT input / Sample Rate: 96 Khz
- In TCNear Patchbay set output Line 1-2 and 3-4 respectively for ADAT 1-2 and ADAT 3-4
- Disconnected Impact Twin from computer.
- Feed the ADAT optical input with 4 channels at 96Khz. They will come out as analog signals in analog outputs 1 to 4.
Question Is it possible to use the Impact Twin as a standalone DA converter for 4 channels ADAT SMUX at 96 Khz Answer This is possible.
Keywords: Impact Twin: Can I use it as a standalone 96 kHz 4-channel DA converter?